The requirements for on-site installations:
The GRP™ series of modules is one of the most compact on the market.
Please note the dimensions from the drawing on the left. Additional 100 mm of space are required in the back of the modules. Each additional valve module will add 73 mm to the height of the system.
Electrical and electronic connections
Electrical and data transfer cables require a free and accessible port or feed through.
1 electrical cable for the synthesis module (8 mm Ø – plug: 30 mm Ø)
1 data transfer cable from synthesis module to notebook (cable: 5 mm Ø – plug: 15×15 mm²)
For mass flow controller: Nitrogen (4 ± 0.5 bar), optionally other gases usable (please contact SCINTOMICS before connection).
Suitable adapters to connect tubing (O.D. 3.2 mm) to the nitrogen outlet and a gauge need to be provided. For reactor cooling, if required: Nitrogen or compressed air (4 ± 0.5 bar)
Suitable adapters to connect tubing (e.g. polyurethane, O.D. 6mm, ID 4mm) to the gas outlet and a gauge need to be provided.
Suitable tubing, gauges and adapters can be provided on request.
Additional space outside the hot cell is needed for a standard 15” Laptop PC and a 24V power supply (90 x 252 x 43.8 mm³ (WxDxH)).
An electrical outlet (100-240 VAC (50/60 Hz), 16A) outside the hot cell is needed to plug in the laptop and the power supply.
Alternatively, if an electrical outlet (100-240 VAC (50/60 Hz), 16A) is available inside the hot cell, the power supply may be placed inside.
Power extension cables
can be provided on request.
For more detailled information on the On-Site Requirements, give us a call.